FEG shuts down randomly without error message.
Titan Cubed G2 60-300
These are possible resolutions for the issue:
- Repair or replace the damaged cables.
These are possible causes for the issue:
- FIB/H cables were damaged due to arcing.
Based on the log file, there are a lot of CPBF communications errors being lost and restored.
While FSE was replacing CPB board, he found some damage on the cable between FIB/H X104 and MWB X10 and he also found burnt marks on FIB/H X11 cable.
Issue seems to be solved after FSE repaired the cable on site but it is advisable to replace the damaged cables if possible.