Some JEOL Microscope Stage LED's cause High Detector Resets & Poor Detector Resolution
- Noran System 6
- Noran System 7
- Contact JEOL to inquire about the possible need for the JEOL installation of the JEOL option "R-Products". This option is totally controlled, installed, and supported by the OEM, JEOL. This option/modification disables the JEOL Stage LED's when the stage is stationary.
- The LED's are active ONLY when the stage is moving. After a stage move, the LED's are disabled again until the next stage move request.
- LED illumination degrade the performance of Thermo's Microanalysis X-Ray detectors causing high resets and poor resolution.