How do I replace the TurboFlow column?
NOTE: Use the Maintenance dialog box to track injection numbers.
1. Turn off the column heater and allow the column heater to cool to room temperature.
a. Open the Aria MX Direct Control window.
b. Click a column heater that appears under the appropriate channel name.
c. To turn off the column heater, click the Enable button so that it shows dark green.
d. Repeat step b and step c to turn off the second column heater.
2. When the column heater has cooled, unwrap the column heater from the column.
3. Replace the column as directed.
Note: Use only your fingers to manipulate the tubing fittings on Prelude systems. Transcend systems require tools, depending on the procedure.