Tracefinder not setting LC and MS to standby
Tracefinder is not setting the LC and MS to standby ONLY when a whole batch or more than one sample is submitted but it will set the LC and MS to standby when only one sample is submitted.
- Mass Spectrometry
Go to TraceFinder configuration and verify that Full sequence submission is selected in the Acquisition submission options.
|Tracefinder acquisition submission options.png||January 10, 2022|