No information appears on display
Autosampler display is showing no information.
UltiMate™ 3000 Autosampler Series:
- WPS-3000(T)SL Analytical
- WPS-3000(T)SL Semi-prep
- WPS-3000TBSL Analytical
|Test Result||Possible Cause||
|No information appears on the autosampler display||
The autosampler is not connected to the mains.
The power is turned off.
The autosampler is in standby mode.
The brightness or contrast is not adjusted correctly.
The fuses blow.
Replacement fuse blows immediately.
An error occurred in the electronic system.
Connect the power cord.
Turn on the power.
Press the Standby key on the front panel.
Adjust the brightness and contrast.
Replace the fuses.